Workers using an implantable cardioverter defibrillator (ICD) are classified by European Directive 2013/35/EU as being at particular risk because of the potential interference between implanted medical devices and electromagnetic fields. The aim of the study was to investigate ICD function using a human-shaped phantom in high magnetic fields of a shunt reactor at a 400 kV substation. We used the phantom in the following experiment periods: isolated from the ground, grounded by a foot, or grounded by a hand. We performed five ICD tests using five different ICD devices. In experiment place A, the magnetic field was over 1000 μT, and in experiment place B, the exposure was over 600 μT. We did not find any disturbances in the ICDs. However, we conducted only 5 ICD experiments in real exposure situations at 400 kV substations. Although it is not possible to draw a strong conclusion regarding risk level, the risk of such ICD disturbances from magnetic field exposure at 400 kV substations does not appear to be high.
2. ANSI/AAMI PC69, Active implantable medical devices --- Electromagnetic compatibility --- EMC test protocols for implantable cardiac pacemakers and implantable cardioverter defibrillators, 81 American National Standard, Association for the Advancement of Medical Instrumentation, Arlington, 2007.
3. European Parliament and Council of the European Union, "Directive 2013/35/EU of the European Parliament and of the council on the minimum health and safety requirements regarding the exposure of workers to the risks arising from physical agents (electromagnetic fields) (20th individual directive within the meaning of Article 16(1) of directive 89/391/EEC) and repealing directive 2004/40/EC," Official Journal of the European Union, 21, 2013.
4. Tiikkaja, M., et al., "Electromagnetic interference with cardiac pacemakers and implantable cardioverter defibrillators from low-frequency electromagnetic fields in vivo," Europace, Vol. 15, No. 3, 388-394, 2013.
5. Katrib, J., M. Nadi, D. Kourtiche, I. Magne, P. Schmitt, M. Souques, and P. Roth, "In vitro assessment of the immunity of implantable cardioverter-defibrillators to magnetic fields of 50/60 Hz," Physiol. Meas., Vol. 34, No. 10, 1281-1292, 2013.
6. Souques, M., I. Magne, and J. Lambrozo, "Implantable cardioverter defibrillator and 50-Hz electric and magnetic fields exposure in the workplace," Int. Arch. Occup. Environ. Health, Vol. 84, 1-6, 2011.
7. Gimbel, J. R. and J. W. Cox, "Electronic article surveillance systems and interactions with implantable cardiac devices: Risk of adverse interactions in public and commercial spaces," Mayo. Clin. Proc., Vol. 82, No. 3, 318-322, 2007.
8. Tiikkaja, M., A. Aro, T. Alanko, H. Lindholm, and M. Hietanen, "Inappropriate implantable cardioverter-defibrillator magnet-mode switch induced by a laptop computer," Pacing Clin. Electrophysiol., Vol. 35, e177-e178, 2012.
9. Korpinen, L., H. Kuisti, J. Elovaara, and V. Virtanen, "Cardiac pacemakers in electric and magnetic fields of 400-kV power lines," Pacing Clin. Electrophysio., Vol. 135, 422-430, 2012.
10. Korpinen, L., H. Kuisti, J. Elovaara, and V. Virtanen, "Implantable cardioverter defibrillators in electric and magnetic fields of 400 kV power lines," Pacing Clin. Electrophysiol., Vol. 37, 297-303, 2014.
11. EN 45502-2-1, "Active implantable medical devices --- Part 2-1: Particular requirements for active implantable medical devices intended to treat bradyarrhytmia (cardiac pacemakers),", 95s, 2003.
12. CENELEC, Low profile connector for implantable cardiac pacemakers, EN 50077, 16, Brussels, Belgium, 1993.