A 132 GHz gyrotron, operating at fundamental harmonic, is designed for the 200 MHz DNP-NMR experiment. In this article, the design of high quality electron beam source is presented. 2.5 dimensional code EGUN and 3 dimensional code CST-Particle Studio are used in the design and optimization of electron gun. The design of electron beam source is performed for a band of magnetic field values at the emitter surface and cavity center which is necessary for the frequency tunabilty of 2-3 GHz needed in DNP/NMR experiments. The results confirm the axial and transverse velocity spreads around 1% and 2.2% and a pitch factor of 1.5. The parametric analyses are also performed for the various electrical parameters such as emitter voltage, anode voltage, emitter magnetic field, etc.
2. Thumm, M., "State-of-the-art of High Power Gyro-devices and Free Electron Masers ," FZK, KIT, 2012.
3. Gaponov, A. V., M. I. Petelin, and V. K. Yulpatov, "The induced radiation of excited classical oscillators and its use in high frequency electronics ," Radiophys. Quantum Electron., Vol. 10, 794-813, 1967.
4. Flyagin, V. A., A. V. Gaponov, I. Petelin, and V. K. Yulpatov, "The gyrotron," IEEE Trans. on Microwave Theory and Tech. , Vol. 25, 514-521, 1977.
5. Chu, K. R., "The electron cyclotron maser --- Relativity in action," Review of Modern Physics, Vol. 76, 489-540, 2004.
6. Kumar, N., U. Singh, T. P. Singh, and A. K. Sinha, "A review on the applications of high power, high frequency microwave source --- Gyrotron," J. of Fusion Energy, Vol. 30, 257-276, 2011.
7. Thumm, M., "Novel applications of millimeter and submillimeter wave gyro-devices," Int. J. of Infrared, Millimeter and Terahertz Wave, Vol. 22, 377-386, 2001.
8. Nanni, E. A., A. B. Barnes, R. G. Griffin, and R. J. Temkin, "THz dynamic nuclear polarization NMR," IEEE Tr. Terahertz Science and Technology, Vol. 1, 145-163, 2011.
9. Bratman, V., M. Glyavin, T. Idehara, Y. Kalynov, A. Luchinin, V. Manuilov, S. Mitsudo, I. Ogawa, T. Saito, Y. I. Tatematsu, and V. Zapevalo, "Review of subterahertz and terahertz gyrodevices at IAP RAS and FIR FU ," IEEE Trans. on Plasma Sci., Vol. 37, 36-43, 2009.
10. Maly, T., G. T. Debelouchina, V. S. Bajaj, K. N. Hu, C. G. Joo, M. L. Mak-Jurkauskas, J. R. Sirigiri, P. C. A. V. D. Wel, and J. Herzfeld, "Dynamic nuclear polarization at high magnetic fields," J. Chem. Physics, Vol. 128, 052211, 2008.
11. Bajaj, V. S., C. T. Farrar, M. K. Hornstein, I. Mastovsky, J. Vieregg, J. Bryant, B. El ena, K. E. Kreischer, R. J. Temkin, and R. G. Griffin , "Dynamic nuclear polarization at 9T using a novel 250 GHz gyrotron microwave source," J. Magnetic Resonance, Vol. 160, 85-90, 2003.
12. Kartikeyan, M. V., E. Borie, and M. K. A. Thumm, Gyrotrons-high Power Microwave and Millimeter Wave Technology, Springer-Verlag, Berlin, Germany, 2004.
13. Edgcombe, C. J., "Gyrotron Oscillators: Their Principles and Practice," Taylor & Francis, 1993.
14., EGUN, (Hermannsfeldt, W. B., Stanford Linear Accelerator Center) , Stanford University Report SLAC-226, 1979.
15. Singh, U., A. Bera, R. R. Rao, and A. K. Sinha, "Synthesized parameters of MIG for 200 kW, 42 GHz gyrotron," J. of Infrared, Millimeter, and Terahertz Waves, Vol. 31, 533-541, 2010.
16. Baird, J. M. and W. Lawson, "Magnetron injection gun (MIG) design for gyrotron applications," Int. J. Electronics, Vol. 61, 953-967, 1986.
17. Lawson, W., "MIG scaling," IEEE Trans. on Plasma Sci., Vol. 16, 290-295, 1988.
18. Tsimring, S. E., "Gyrotron electron beams: Velocity and energy spread and beam instabilities," Int. J. of Infrared, Millimeter and Terahertz Wave, Vol. 22, 1433-1468, 2001.
19., "User Manual: 2010 Version of CST, CST Particle Studio GmbH, Darmstadt,".
20. Singh, U., A. Bera, N. Kumar, L. P. Purohit, and A. K. Sinha, "Three-dimensional simulation of MIG for 42-GHz 200-kW gyrotron," IEEE Trans. on Plasma Sci., Vol. 38, 1546-1550, 2010.