This paper presents an experimental study of the performance of a wireless resonant energy link for implantable biomedical devices. More specifically, the proposed system consists of two planar resonators: a primary resonator that is connected to a power source and operates outside the body, and a secondary resonator that is connected to the implanted device and operates inside the body. Each resonator is a planar spiral resonator; the wireless power transmission is obtained by exploiting the magnetic coupling between the two resonators when they are operating at small distances. A prototype working in the ISM band centered at 434 MHz has been developed and analyzed. Reported results confirm that the proposed system is a viable solution for wirelessly providing implantable devices with the power necessary for operation.
2. Park, D. J., Y. J. Lee, and J. Y. Park, "Long-term stabled non-enzymatic glucose sensor for continuously monitoring system applications," Proc. of the IEEE Nano/Micro Engineered and Molecular Systems, 704-707, China, 2008.
3. Salam, M. T., D. K. Nguyen, and M. Sawan, "A low-power implantable device for epileptic seizure detection and neurostimulation," Proc. of IEEE Biomedical Circuits and Systems Conference (BioCAS), 154-157, Paphos, Cyprus, 2010.
4. Vidal, N., S. Curto, J. M. Lopez-Villegas, J. Sieiro, and F. M. Ramos, "Detuning study of implantable antennas inside the human body ," Progress In Electromagnetics Research, Vol. 124, 265-283, 2012.
5. Gemio, J., J. Parron, and J. Soler, "Human body effects on implantable antennas for ism bands applications: Models comparison and propagation losses study," Progress In Electromagnetics Research, Vol. 110, 437-452, 2010.
6. Ashoori, E., et al., "Design of double layer printed spiral coils for wirelessly-powered biomedical implants," Proc. of Engineering in Medicine and Biology Society, 2882-2885, Boston,Massachusetts, 2011.
7. Goto, K. , T. Nakagawa, and S. Kawata, "An implantable power supply with an optically rechargeable lithium battery," IEEE Trans. Biomed. Eng., Vol. 48, No. 7, 830-833, 2001.
8. Huang, F. J., et al., "Rectenna application of miniaturized implantable antenna design for triple-band biotelemetry communication," IEEE Trans. on Antennas and Propagation, Vol. 59, No. 7, 2646-2653, 2011.
9. Laskovski, A. N. , M. R. Yuce, and T. Dissanayake, "Stacked spirals for biosensor telemetry," IEEE Sensor Journal, Vol. 11, No. 6, 1484-1490, 2011.
10. Jow, U. M. and M. Ghovanloo, "Modeling and optimization of printed spiral coils in air, saline, and muscle tissue environments," IEEE Trans. on Biomedical Circuits and Systems, Vol. 3, No. 5, 339-347, 2009.
11. Jung, K. H., et al., "Wireless power transmission for implantable devices using inductive component of closed magnetic circuit," Electronics Letters, Vol. 45, No. 1, 21-22, 2009.
12. Kumar, A. , S. Mirabbasi, and M. Chiao, "Resonance-based wireless power delivery for implantable devices," Proc. of IEEE Biomedical Circuits and Systems Conference, 25-28, Beijing, China, 2009.
13. Tesla, N., "Apparatus for transmitting electrical energy,", U.S. Patent, 1119732, 1914.
14. Monti, G. and F. Congedo, "UHF rectenna using a bowtie antenna," Progress In Electromagnetics Research C, Vol. 26, 181-192, 2012.
15. Monti, G. , L. Tarricone, and M. Spartano, "X-band planar rectenna," Antennas and Wireless Propagation Letters, Vol. 10, 1116-1119, 2011.
16. Vidal, N. , S. Curto, J. M. Lopez-Villegas, J. Sieiro, and F. M. Ramos, "Detuning study of implantable antennas inside the human body ," Progress In Electromagnetics Research, Vol. 124, 265-283, 2012.
17. Gemio, J., J. Parron, and J. Soler, "Human body effects on implantable antennas for ism bands applications: Models comparison and propagation losses study," Progress In Electromagnetics Research, Vol. 110, 437-452, 2010.
18. Li, X. and L. Cao, "Microstrip-based segmented coupling reader antenna for near-field RFID applications," Microwave and Optical Technology Lett., Vol. 53, No. 8, 1774-1777, 2011.
19. Qing, X. and Z. N. Chen, "Segmented spiral antenna for UHF near-field RFID," Proc. of IEEE Int. Symp. on Antennas and Propagation APSURSI, 996-999, 2011.
20. Poon, , A. S. Y. , S. O'Driscoll, and T. H. Meng, "Optimal operating frequency in wireless power transmission for implantable devices," Proc. of the 29th Int. Conf. of the IEEE Eng. in Medicine and Biology Society, Lyon, France, Aug. 23-26, 2007.
21. Gabriel, C., S. Gabriel, and E. Corthout, "The dielectric properties of biological tissues: I. Literature survey," Phys. Med. Biol., Vol. 41, 2231-2249, 1996.
22. Gabriel, S., R. W. Lau, and C. Gabriel, "The dielectric properties of biological tissues: II. Measurements in the frequency range 10 Hz to 20 GHz," Phys. Med. Biol., Vol. 41, 2251-2269, 1996.
23. Gabriel, S., R. W. Lau, and C. Gabriel, "The dielectric properties of biological tissues: III. Parametric models for the dielectric spectrum of tissues," Phys. Med. Biol., Vol. 41, 2271-2293, 1996.
24. Oh, J. H., T. H. Kim, J. H. Yoo, J. K. Pack, Y. M. Yoon, M. Y. Choi, and S. Y. Lee, "Human exposure assessment for wireless power transmission system," PIERS Proceedings, Xi'an, China, Mar. 22-26, 2010..