A Transverse Electromagnetic Mode (TEM) cell is one interesting alternative for studies of biological effects of radiofrequency radiation at reduced scale (in vitro studies). Controlled and well-characterized exposure conditions are essential for a concluding investigation: the biological sample has to be exposed to a uniform incident electromagnetic wave and the dose of absorbed radiation has to be precisely determined and correlated with the effect. Unfortunatelly, many times experimental dosimetry is either unavailable or unappliable, so that a pre-characterised and validated experimental set-up is mostly valuable. In this regard, the main objective of present work was to experimentally validate a computational model of an own-built TEM cell designed for bioelectromagnetic experiments in frequency range of 100MHz-1GHz. For validation, three significant parameters were investigated comparatively, by measurements and by computation: scattering parameters; incident electric field distribution; absorbed power in a set of liquid samples. By using the finite integration technique (FIT) method implemented by the commercial code CST Microwave Studio, and by using a vector network analyzer in the experimental approach, we validated the designed TEM cell and characterized it successfully. The second objective was a dosimetric study of four different liquid samples loaded in the cell. We used the absorption coefficient (AC) which may be assimilated to the specific absorption rate (SAR) of energy deposition in the whole sample volume. AC was shown to converge in experiment and simulation up to 800MHz for all samples. AC didn't depend directly upon sample's volume (even if, frequently, greater volumes showed higher absorption) but rather upon the internal field distribution in the sample, distribution that mostly depends on the frequency and on the dimensions of the liquid samples.
2. Durney, C. H., H. Massoudi, and M. F. Iskander, Radiofrequency Radiation Dosimetry Handbook, 4th Ed., USAFSAM-TR-85-73, Brooks AFB, Texas, USA, 1986, http://www.radhaz.com/docs/RF%20Dosimetry%20Handbook%20ver%204.pdf .
3. Guy, A. W., C. K. Chou, and J. A. McDougall, "A quarter century of in vitro research: A new look at exposure methods," Bioelectromagnetics, Vol. 20, No. S4, 21-39, 1999.
4. Tippet, J. C. and D. C. Chang, "Radiation characteristics of electrically small devices in a TEM transmission cell," IEEE Transactions on Electromagnetic Compatibility, Vol. 18, No. 4, 134-140, 1976, http://www.jpier.org/PIERB/pierb29/13.11022506.pdf.
5. Iftode, C., S. Miclaus, P. Bechet, and E. Surducan, A TEM cell model analysis for radiofrequency dosimetry improvement by computational means , Proceedings of the 7th International Symposium on Advanced Topics in Electrical Engineering, Bucharest, May 12-15, 2011.
6. Burkhardt, M., K. Pokoviic, M. Gnos, T. Schmid, and N. Kuster, "Numerical and experimental dosimetry of Petri dish exposure setups," Bioelectromagnetics, Vol. 17, No. 6, 483-493, 1996.
7. Green, H. E., "The TEM-mode bandwidth of two-conductor open transmission lines," Progress In Electromagnetics Research, Vol. 40, 1-28, 2003.
8. Nikoloski, N., J. Fröhlich, T. Samaras, J. Schuderer, and N. Kuster, "Reevaluation and improved design of the TEM cell in vitro exposure unit for replication studies," Bioelectromagnetics, Vol. 26, No. 3, 215-224, 2005.
9. Boriraksantikul, N., P. Kirawanich, and N. E. Islam, "Near-field radiation from commercial cellular phones using a TEM cell," Progress In Electromagnetics Research B, Vol. 11, 15-28, 2009.
10. Lee, S. U., H. J. Eom, and J. H. Kwon, "TEM mode in the GTEM cell," Journal of Electromagnetic Waves and Applications, Vol. 25, No. 4, 519-526, 2011.
11. Calo, G. and V. Petruzzelli, "Electromagnetic and thermal analyses of improved GTEM cells for bioelectromagnetic experiments," Progress In Electromagnetics Research, Vol. 125, 503-526, 2012.
12. Zhao, J. X., H. M. Lu, and J. Deng, "Dosimetry and temperature evaluations of a 1800MHz TEM cell for in vitro exposure with standing waves," Progress In Electromagnetics Research, Vol. 124, 487-510, 2012.
13. Rostamzadeh, C., B. Archambeault, and S. Connor, FDTD analysis of symmetric TEM cell, IEEE International Symposium on Electromagnetic Compatibility, Aug. 8-12, 2005.
14. Moglie, F., V. M. Primiani, and A. P. Pastore, "Modeling of the human exposure inside a random plane wave field," Progress In Electromagnetic Research B, Vol. 29, 251-267, 2011.
15. Malaric, K. and J. Bartolic, "Design of a TEM-cell with increased usable test area," Turkish Journal Electrical Engineering, Vol. 11, No. 2, 143-154, 2003.
17. Clemens, M. and T. Weiland, "Discrete electromagnetism with the finite integration technique," Progress In Electromagnetics Research, Vol. 32, 65-87, 2001.
18. Sevat, P. Design of a TEM cell EMP simulator, Report No. 1084, Defence Research Establishment, Ottawa, Jun. 1991.
19. Zhong, C., Examinations of higher order mode cutoff frequencies in symmetrical TEM cells , Proc. of International Symposium on Electromagnetic Compatibility, EMC 2009, 6-11, 2009.
20. Wilson, P. F. and M. T. Ma, "Simple approximate expressions for higher order mode cut-off and resonant frequencies in TEM cells," IEEE Transactions EMC, Vol. 28, No. 3, 125-130, Aug. 1986.
21. Garbe, H. and C. Groh, Calculating the usable frequency range of TEM-waveguides, Proceedings of the 27th General Assembly of the International Union of Radio Science, Maastricht, NL, Aug. 2002.
22. Morega, M., S. Miclaus, and A. Machedon, "Analysis of the electromagnetic field in a controlled enclosure for biological dosimetry ," Rev. Roum. Sci. Techn. --- Électrotechn. et Énerg., Vol. 52, No. 2, 225-235, Bucarest, 2008.
23. Morioka, T., A field uniformity study of a TEM cell by using a short wire scatterer, IEEE International Symposium on Electromagnetic Compatibility, Jul. 9-13, 2007.
24. Islam, M. T., H. Z. Abidin, M. R. I. Faruque, and N. Misran, "Analysis of materials effects on radio frequency electromagnetic fields in human head," Progress In Electromagnetics Research, Vol. 128, 121-136, 2012.
25. Wang, Y. and M. N. Afsar, "Measurement of complex permittivity of liquids using waveguide techniques," Progress In Electromagnetics Research, Vol. 42, 131-142, 2003.
26. Angulo, L. D., S. G. Garcia, M. F. Pantoja, C. C. Sanchez, and R. G. Martin, "Improving the SAR distribution in petri-dish cell cultures," Journal of Electromagnetic Waves and Applications, Vol. 24, No. 5-6, 815-826, 2010.
27. Zhang, M. and A. Alden, "Calculation of whole-body SAR from a 100MHz dipole antenna," Progress In Electromagnetics Research, Vol. 119, 133-153, 2011.
28. Complex Dielectric Constant of Water Calculator, http://www.random-science-tools.com/electronics/water dielectric.htm.
29. Hill, R. Performance tests of indexsar `Di-line' TEM sensors for dielectric property measurement of tissue-simulant liquids for SAR testing, Surrey RH5 5DR, Indexsar, UK, Jul. 2002.
30. Aziz, M. S., A. G. Mostafa, A. M. Youssef, and S. M. S. Youssif, "Electrical conductivity and dielectric properties of bulk glass V2O5 (ZnO, PbO) SrO FeO," Physics Research International, Vol. 2011, 10 pages, 2011, Article ID 583420, DOI:10.1155/2011/583420.
31. Li, M.-H., H.-L. Yang, and X.-W. Hou, "Perfect metamaterial absorber with dual bands," Progress In Electromagnetics Research, Vol. 108, 37-49, 2010.
32. Gangwar, R. K., S. P. Singh, and D. Kumar, "SAR distribution in a bio-medium in close proximity with rectangular dielectric resonator antenna," Progress In Electromagnetics Research B, Vol. 31, 157-173, 2011.
33. ISO/IEC Guide of the expression of uncertainty in measurement, Geneva, 1995.
34. EA Guideline 4/16, "Expression of uncertainty in quantitative testing,", 2003.
35. Laakso, I., "Uncertainty in computational RF dosimetry," Doctoral Dissertation, Aalto University, 2011.